Clean-Trace Water Plus Total ATP Swabs (AQT200) 100 Pack
Available to ship on or before:10/15/2022
Add To Cart
Only ships Next Day Air.
To prevent delay, select Next Day Air at checkout. Since refrigeration is required, we ship swabs Monday through Thursday only (to prevent the parcel from being left in terminal over the weekend), in temperature packs with ice pack included when necessary.
3M Clean-Trace Water Plus Total ATP (AQT200) 100 Pack
Real-time ATP water hygiene test which gives an objective indication of cleanliness in seconds.
- the test is designed to rapidly assess equipment sanitation processes and the efficiency of Clean In Place (CIP) procedures by the improved more accurate detection of organic contamination from microbial or product residues.
- The test gives an objective indication of cleanliness in water samples in seconds. Designed
for use with the 3M™ Clean-Trace™ NG Luminometer, the new AQT200 formulation now
provides the ability to gain more consistent and reliable results providing greater confidence
in your hygiene monitoring procedures.
- The new and improved formulation also provides the flexibility to store AQT200 at your own
convenience - room temperature for up to 2 months or refrigerated for up to 10 months.
- ECCN: EAR99.
- Harmonized Code: 3822005090.
- 3M Stock #: 7100002659.
- Previous 3M Stock # GH6205381123M.
- Catalog #: AQT200.
- UPC #: 04054596181359.
Results measured using the 3M Clean-Trace NG or LM1 Luminometer and provided in Relative Light Units (RLU).
- Optimal test temperature: 59° to 86° F (15° to 30° C)
- Shelf life:
- Refrigerated: 35.6° to 46.4°F (2° to 8°C): approximately 8 months
- Room temperature: 69.8° to 77° F (21° to 25°C): approximately 2 months
Instructions for Use
- Before the 3M Clean-Trace Water-Free test device is activated, it is
important to ensure your 3M Clean-Trace NG or LM1 Luminometer is switched on
and initialized. Refer to the manual provided with the instrument for full
details. Place the 3M Clean-Trace Water-Free ATP test devices at room
temperature for at least 10 minutes before use.
- Collect liquid samples from the test sites of interest. Process each
sample through steps 3-6. DO NOT batch at any of these stages.
- Swirl to mix the sample. Remove a 3M Clean-Trace Water-Free ATP test
device from the foil pouch. Remove the sampling stick from the 3M device
and immerse the sample collection rings into the liquid under test, tapping
the handle gently if bubbles form (figure 1). Immediately remove the
sample head from the solution to be tested and carefully return the
sampling stick to the test device such that the handle is at its starting
- To process the sample, push down firmly on the top of the 3M Clean-Trace
Water-Free ATP sample stick handle (figure 2). The handle will slide into
the test device tube and the top of the handle should be level with the top
of the device tube when fully depressed. Grip the top of the test device
and shake rapidly side-to-side for at least five seconds to mix the
sample and reagent (figure 3).
- Immediately open the sample chamber of the 3M Clean-Trace NG or LM1 Luminometer
and insert the 3M Clean-Trace Water-Free ATP test device. Close the
chamber cap and press the measure button. The light emitted by the 3M Clean-Trace test device will be measured and the result (in RLU) will
appear on the display (figure 4).
Search Category: Business & Industrial